GNOMIX-PVT apparatus (GNOMIX INC., 3809 Birchwoold Drive, Boulder, CO 80304, USA) for detection of changes in spezific volume as function of pressure, temperature and time: Δ∨sp = f(P,t,t) (alternatively also ΔP = f(∨sp,T,t)).
Principle: Confining fluid (usually mercury) technique, ensuring hydrodynamic pressure to the sample at all conditions.
Sample: 1-2 mL, liquid or solid with variable geometry, no fine powder, no gas inclusions.
Measuring range: room temperature to 400 °C, 10 MPa to 200 MPa
Accuracy: +/- 0.002 cm3/g (below 200 °C); +/- 0.004 cm3/g (above 200 °C); changes in ∨sp as small as 0.0002 cm3/g can be resolved reliably.
Powder conductivity measuring unit (self-made): Determination of bulk conductivity of solids (especially powders) by two electrode (D = 5 mm) system under pressure (up to 30 MPa); measuring range of resistance: 1 µΩ to 1GΩ.